2pcs Digital 0.0-14.0 PH Meter Tester 0-9990ppm Digital TDS EC LCD Water Purity PPM Aquarium Filter
Dear customer friend,thanks for your support,pls note that ph meter need LR44 battery and TDS&EC meter need CR2032 battery,because if the product with battery, the customs will not pass, so the product not include the battery, hope you can understand, thank you.
Easy to Use: Simply immerse the pH meter electrode in the solution to be tested, readings are quickly display on a large lcd display
Wide Application: Ideal instrument for any aquarium, the fishing industry, swimming pools, spa, school laboratories, food & beverage, drinking water etc.
Auto Calibrated: Features in-built calibration with 2 buffer powders(included).Just press a button to finish the calibration. This pH meter is factory calibrated. Users can use it directly.
Fast and Accurate Results: No need to work with those annoying color charts and test strips.Resolution: 0.01 pH, Accuracy: ±0.01 pH
Compact Portable Design: Pocket fitting pH meter that’s also lightweight and compact. Total concealed fit in your bag or pocket for easy, casual carrying.
Provides accurate and reliable test readings: 3-In-1 Digital TDS meter provides accurate readings for TDS (Total Dissolved Solids) EC (Electrical Conductivity) and temperature. Professional, Accurate, and Precise 3-in-1 TDS, EC, and Water Temperature Meter. The TDS meter has automatic temperature compensation in measurement mode. 9 minutes no operation, the meter will automatically shut down. Save power, extend battery life
Display: LCD Digital
Operating Temperature: 0~60°C
Measuring Range: 0.00~14.00
Probe Type: Precision glass bulb electrode
Lock function: makes it easy to read and record, press HOLD button then remove from test solution.
2.First rinse the electrode with distilled water and suck it with filter paper.
3.Turn the meter on by pressing “ON/OFF” key.
4.Immerse the pH meter electrode in the solution to be tested (should not be over the immersion line).
5.Stir gently and wait around 30 seconds till the reading stabilizes.
6.After finishing, clear the electrode with pure water, turn the meter off by pressing “ON/OFF” key.
7.Always replace the protective cap after use.
1.Turn on PH meter
2.Disolve each buffer solution in 250ML of distilled water
3.Immerse the electrode into the PH 6.86 solution,(under the temperature of 25)
4.Press the “CAL”(calibration)button for 5 seconds and release
-Display will start flashing 6.86
-Wait until the display stops flashing.
Rinse the electrode with distilled water and dry it with filter paper.
5.Immerse the electrode in PH 4.00 solution,
-Press“CAL”button for 5 seconds,then press and release immediately second time
-Reading will start flashing 4.00
-Wait until display stops flashing
Rinse and dry the electrode with distilled water as before.
6.Place the meter back to the buffer solution to test calibration.If incorrect,please repeat the calibration
7.If approximate PH of your test solution is known to be above 7.0,calibrate the meter using 6.86 and 9.18 solution. To do this,please follow the calibration chart,The third step is the calibration method for 9.18 point.